Frequently Asked Questions for SmartReader Plus 4, and SmartReader Plus 4 LPD
No. The media should be non-corrosive gases and dry air.
A: Gauge pressure is Zero PSI at atmospheric pressure.
Absolute pressure includes the atmospheric pressure. So, typically with no pressure applied, it will read approximately 14.7 PSI.
Absolute pressure = Gauge pressure + atmospheric pressure (for example: if the atmospheric pressure where you are is 14.7 PSI and your gauge pressure is 1 PSI then the Absolute pressure will be 15.7 PSI).
You can use SRP-004-30A or SR-004-30A loggers.
The accuracy of the pressure channels is +/-1% of full scale at 25°C (77°F) for the SmartReader 4 logger and +/-0.5% of full scale for the SmartReader Plus 4. This only applies over the range of 0-70°C (32 to 160°F), as the pressure modules require temperature compensation.
The maximum pressure range for the SRP4 is 150PSI Gauge. If you have a pressure transducer with the output of 0-25 mADC, or 0-200 mVDC, 0-2.5, 5, 10 Volt DC, then you can connect it to the SR7 or SRP7 (process signal logger). You can also write a custom equation to convert the unit and get the pressure readings in the required units (e.g. PSI).
It is a Silicon piezoresistive strain gauge.
±0.05”WC (±12.5Pa) for the ±2”, ±1” & ±0.5”WC ranges; ±0.1”WC (±25Pa) for the ±5” & ±10”WC ranges.